- Technical Service
- Service Promotions
- Videos & Tutorials
- Hygiene & Maintenance
- Service Station Locator
- Disposal Guidelines
Was this information helpful? Help us to improve our service.Contact
Resolve problems quickly and simply yourself. Using the product-specific troubleshooting, you can find the solution quickly here.
During a vacuum phase of the cycle, a vacuum of 0.20 bar or lower could not be archieved.Action:
Clean, check or replace the door seal (see 11.2 of your Instructions for Use Manual or the instrumental video "Changing the door seal") and the face side of the chamber; repeat the cycle. If the problem pesists, call your W&H Premium Service Partner.